### Aortic valve area calculation

Aortic valve area calculation is an indirect method of determining the area of the aortic valve. The calculated aortic valve orifice area is currently one of the measures for evaluating the severity of aortic stenosis. A valve area of less than 0.8 cm² is considered to be severe aortic stenosis.There are many ways to calculate the valve area of aortic stenosis. The most commonly used methods involve measurements taken during echocardiography. For interpretation of these values, the area is generally divided by the body surface area, to arrive at the patient's optimal aortic valve orifice area.

### Planimetry

Planimetry is the tracing out of the opening of the aortic valve in a still image obtained during echocardiographic acquisition during ventricular systole, when the valve is supposed to be open. While this method directly measures the valve area, the image may be difficult to obtain due to artifacts during echocardiography, and the measurements are dependent on the technician who has to manually trace the perimeter of the open aortic valve. Because of these reasons, planimetry of aortic valve is not routinely performed.

### The continuity equation

The continuity equation states that the flow in one area must equal the flow in a second area if there are no shunts in between the two areas. In practical terms, the flow from the left ventricular outflow tract (LVOT) is compared to the flow at the level of the aortic valve. Using Echocardiography, the aortic valve area calculated using the time velocity integral (TVI) is most accurate method and is the preferred method. The flow through the LVOT, or LV Stroke Volume (cm3 or cc), can be calculated by measuring the LVOT diameter (cm), squaring that value, multiplying the value by 0.78540 giving cross sectional area of the LVOT (cm2)and multiplying that value by the LVOT TVI (cm), measured on the spectral Doppler display using pulsed-wave Doppler. From these, it is easy to calculate the area (cm2) of the aortic valve by simply dividing the LV Stroke Volume (cm3) by the AV TVI (cm) measured on the spectral Doppler display using continuous-wave Doppler.$Aortic\ Valve\ Area \left(cm^2\right)=\left\{diameter^2$
• 0.78540
• LVOT\ TVI /Aortic Valve\ TVI}
• The weakest aspect of this calculation is the variability in measurement of LVOT area, because it involves squaring the LVOT dimension. Therefore, it is crucial for the sonographer to take great care in measuring the LVOT diameter.

### The Gorlin equation

The Gorlin equation states that the aortic valve area is equal to the flow through the aortic valve during ventricular systole divided by the systolic pressure gradient across the valve times a constant. The flow across the aortic valve is calculated by taking the cardiac output (measured in liters/minute) and dividing it by the heart rate (to give output per cardiac cycle) and then dividing it by the systolic ejection period measured in seconds per beat (to give flow per ventricular contraction).$Valve\ Area\ \left(cm^2\right)=\frac\left\{Cardiac\ Output \left(\frac\left\{ml\right\}\left\{min\right\}\right)\right\}\left\{Heart\ rate\ \left(\frac\left\{beats\right\}\left\{min\right\}\right)\cdot Systolic\ ejection\ period\ \left(s\right)\cdot 44.3 \cdot \sqrt\left\{mean Gradient\ \left(mmHg\right)$The Gorlin equation is related to flow across the valve. Because of this, the valve area may be erroneously calculated as stenotic if the flow across the valve is low (ie: if the cardiac output is low). The measurement of the true gradient is accomplished by temporarily increasing the cardiac output by the infusion of positive inotropic agents, such as dobutamine.

### The Hakki equation

The Hakki equation is a simplification of the Gorlin equation, relying on the observation that in most cases, the numerical value of $heart rate \left(bpm\right) \cdot systolic ejection period \left(s\right) \cdot 44.3 \approx 1000$. The resulting simplified formula is:$Aortic\ Valve\ area\ \left(cm^2\right)\approx\frac\left\{Cardiac\ Output\ \left(\frac\left\{litre\right\}\left\{min\right\}\right)\right\}\left\{\sqrt\left\{peak Gradient\ \left(dll\right)$

Next Page